A promising tool for surgical lipotransfer: a constant pressure and quantity injection device in facial fat grafting
© The Author(s) 2017
Received: 15 January 2016
Accepted: 24 March 2017
Published: 31 May 2017
The purpose of this study was to preliminarily assess a constant pressure and quantity fat granule injection device for minimal invasive properties in clinic.
A retrospective controlled study was carried out, from October 2013 to January 2015, on 76 female healthy patients aged between 26 and 53 years at the General Hospital of Lanzhou Military Command, China. To achieve small volume, high thrust, and precision requirements of facial fat grafting, an integrated handheld controller and motor design was adopted, reducing the volume and weight of the fat transplantation injection device. The 76 patients underwent 90 procedures each side of the face; each patient was treated with the aforementioned device on the face’s left side, while a conventional hand-push injection device was used on the right side as control. The outcome was assessed on pre- and postoperative images, with 6-24 months follow-up.
The current device consistently allowed deposition of fat threads at about 55 μL/cm after cannula withdrawal; the volume of fat injected could be precisely adjusted to 0.04 mL/s. This device had the advantages of small-volume injection and convenient operation. The patients exhibited a good, stable shape and a smooth contour line in both sides. The long term satisfaction was higher for the left side than for the right one. Nodules and unevenness occurred only on the right side. Ecchymosis occurred significantly less frequent on the left side than the right one. Intraoperative pain was significantly lower for the left side than the right one.
This device offered superior control compared with the conventional one and constitutes a promising tool for surgeons practicing lipotransfer.
Autologous fat grafting has helped for more than a century treat various conditions, such as syndromic defects, posttraumatic and postoperative complications, velopharyngeal insufficiency, and esthetic disorders, with the technique gaining increasing popularity in the last 25 years . However, no consensual approach exists as for the best technique to be used. Various aspects of fat grafting have been studied, including covering the donor site, aspiration methods, local anesthesia, fatty tissue processing, reinjection, and graft stability. Unfortunately, many proposed techniques were not validated in well-designed clinical trials. Meanwhile, the final outcomes evaluated usually lack quantitation of fat grafting viability [1, 2].
Carpaneda et al.  found 40% graft survival at 1.5 ± 0.5 mm around the graft margin, with thickness and shape determining fat transplantation success; they proposed to use fat grafts with diameters < 3 mm for high graft survival . Coleman  conceived the notion of structure fat grafting, insisting fat parcels ought to be manually placed with < 1/10 mL injection volume per parcel.
To enhance grafting survivability, small parcels of fat must be distributed evenly throughout the recipient tissue; the parcels should be small enough for plasmatic imbibition to occur, thereby providing sufficient nutrition to support the survival of fat grafts until neovascularization develops. Therefore, injection equipment and techniques are particularly important in optimizing the clinical effects .
To consistently inject a fat thread at every cannula passage, a constant pressure and quantity injection device was developed. The device releases a set volume of filler material per unit distance after cannula withdrawal. The current study aimed to describe the architectonic of the injection device, which was also assessed as an automated tool in facial fat grafting.
The front part of the device consists of a bionic design rack, which allows a firm grip by the operator. The rear portion has a 3.6 V rechargeable lithium battery, which can be used repeatedly. The device is made of stainless steel and acrylic materials, is reusable, and uses cobalt 60 as a source for radiation disinfection or low temperature plasma sterilization.
The device injects 55 μL of filler material, and cannula withdrawal from tissue was performed every 1 cm. The amounts of fat injected can be accurately adjusted to 0.04 mL/s, somewhat below the internal volume of a 12-gauge cannula. Consistent fat threads were produced in the recipient bed thanks to uniform distribution.
A retrospective controlled study was carried out from October 2013 to January 2015 on 76 healthy female patients aged between 26 and 53 years at the General Hospital of Lanzhou Military Command, China. Approval was obtained for the current study from the hospital’s Ethics Committee. All patients who underwent facial fat grafting provided informed consent allowing the device described below to be used.
The device was not used in the following situations: chemotherapy, radiotherapy, prolonged steroid use, connective tissue or chronic blood ailments, and systemic metabolic disorders. The patients were also excluded with asymmetric facial features.
A tumescent technique with a 10-mL syringe under low-pressure suction harvest was employed. The abdominal area was most commonly employed as the donor site. Other sites included thighs or buttocks. The syringe was kept upright until fat separation to yield a distinct yellow layer, which was collected. The wicking method was employed to remove oil and nonviable fat. The collected fat was then sampled into 1 mL syringes for injection.
Additional file 1: The autologous fat transplantation for frontal. (MP4 1618 kb)
Images were acquired before operation, right after surgery, and at 6-24 postoperative months. Position, facial expression, focal distance, and camera configurations were all standardized. Patients, plastic surgeons, and a third party unrelated to the study were required to evaluate the therapeutic effects, which included: (1) general level of satisfaction for each side; (2) notable asymmetry reflected by treated area volumes; (3) nodules, unevenness, ecchymosis, and pain; (4) skin texture amelioration.
Chi-square test and Fisher's exact test were employed to statistically assess the aforementioned points. Significance level was set at P < 0.05.
Comparison of two groups of patients for treatment evaluation n (%)
Left side (using the device) n(%)
Right side (using the conventional technique) n(%)
1. General level of satisfaction
2. Notable asymmetries in terms of volume
3. Notable irregularities
4. Level of improvement in skin texture
Autologous fat grafting is commonly used for correcting anomalies (contours) in birth defects, trauma sequelae, surgery rejuvenation, and burn injuries. Burn sequelae result in local volume loss and skin scarring. Autologous fat grafting can restore a missing relief by filling a localized depression, reshaping the face volume, or smoothing a scarring skin [9, 10]. Although autologous fat transfer usually generates optimism in clinic, there is uncertainty concerning the graft viability .
In the past decades, fat-grafting procedures have undergone important changes to improve fat survival or retention. Gonzalez et al.  advocated the gap formed by a puncture prior to injection; Barret et al.  and Coleman  proposed multichannel, multipoint, and small amount injections using a 1-mL or 5-mL syringe or thin sleeve; Xie et al.  developed the “3L3M” autologous fat transplantation technique. Adopting a multilayer, multiway tunnel that can improve the survival rate has become the consensual view. Injecting the same small volume of fat is very important. Too much would kill fat cells and would cause inflammation and edema with inconsistent outcomes. The conventional hand-push injection method hardly controls the injection amounts and prolongs postoperative edema. In addition, it could cause needle- and cannula-associated problems of clogging. This leads to the surgeon increasing the syringe pressure, which not only enhances the fat amounts administered to unintended areas, but also can cause damage to fat cells . Moreover, it is difficult to administer a consistent thread of fat after cannula withdrawal; therefore, fat deposits tend to form clumps, with irregular distribution and subsequent necrosis. Because of fine anatomic and complex morphological structure, facial appearance requirements are higher. Considering the stereo frame structures of the dermis, ligaments, and muscles, a scientific comparison of the fat fine grafting technique and grafting tools is needed.
To inject consistent threads of fat at every cannula passage, multiple innovative techniques have been proposed for ameliorating fat grafting [15–20]. Fulton et al.  attached the syringe to the scale of the fat grafting gun and pulled the trigger once the syringe pushed forward a grid (of 0.1 ml); although accurate, the injection quantity was too high. Yoshimura et al.  separated the syringe from the needle, had the syringe pushed by a specialized doctor while other physicians controlled the needle’s movement, so as to achieve constant force and pressure injection. Hetherington et al.  invented a prototype injection control device driven by a motor using a series of transmission gears to precisely control the speed and quantity of injection. In recent years, the commercially available micro-autologous fat transplantation (MAFT) device invented by Lin et al.  has been employed for fat grafting. MAFT’s advantage is to convey fat parcels of accurately controlled volumes per injection. However, this invention was based on the burst mode in which the operator needs to constantly pull the trigger, resulting in fatigue in fingers, human consumption, and bolus incoherence. Moreover, in some burn sequelae, tight local fibrous adhesions will increase the injection pressure; the burst mode is affected by the interference of pressure, and consistent fat parcel deposition cannot be achieved.
To avoid these shortcomings, the current study independently developed a constant pressure, constant quantity, and minimally invasive fat transplantation injection device. In the newly engineered device, an electric drive was adopted through a gear motor drive reducer, with a screw rotation drive pushing the syringe handle forward at a constant speed. The electric drive is not subject to pressure interference during injection, as it can continue to maintain a constant speed. A current limiting device was included in the circuit board with self-protection effects, to prevent injecting the bolus too fast, so as to maintain a uniform bolus. This injection device can dispense extremely small and uniform threads of fat with high precision, achieving constant injection and minimizing tissue trauma. With 1-mL syringes, it injects 56 μL of fat granules every centimeter after cannula withdrawal. Using this tool, uniform thread-shaped fat is always delivered into the recipient bed. Also, variable tissue resistance would not affect fat injection since the device is equipped with an electric drive.
The proposed device prevents tissue trauma. With equal fat amounts administered, the volume of fat in every unit length tunnel is small; this results in low local tissue tension, therefore, decreasing pain sensation. This technique has shifted the focus of fat injection from operation skills to accurate and quantified fat delivery, thereby making fat grafting easy, efficient, and timesaving.
In this study, the minimum injection site area was about 1.0 cm2, with a minimum filling volume of only 0.05 mL. It is difficult for a conventional hand-push injection, which relies solely on the surgeon’s experience. to control the filling amounts and to inject so limited fine fat filling. This device can be used to precisely control the filling volume, shorten operative time, and reduce tissue damage and bleeding, thereby, significantly reducing pain and reducing the surgeon’s efforts.
Here, one half of the face was treated with the developed device, while the other underwent a conventional hand-push injection. As shown above, the clinical effectiveness of the current device was superior to that of the syringe-push mode; nodules, unevenness, ecchymosis, and pain were significantly reduced, with higher satisfaction of patients at clinical follow-up. Because of these findings, the authors adopted this fat transplantation injection device after years of using the conventional hand-push injection.
The constant pressure and quantity injection device proposed in this study yields higher overall satisfaction than the conventional hand-push injection tool. More consistent and clinical esthetic results, faster recovery, and decreased operation time were observed in comparison with conventional hand-push injection. One may therefore conclude that this device is important for lipotransfer.
This study was supported by grants from the key project of “12th five-year plan” for medical science and technology of PLA (no. BWS11C061) and the National Natural Science Foundation of China (grant no. 81401612). The device has been authorized by China Patent Bureau (ZL201420213962.0).
Availability of data and materials
The datasets used and analyzed during the current study are available from the corresponding author on reasonable request.
YL and MS conceived and designed the study. YL, MS, PL, and XZ performed the experiments. YL, MS, PL, and XZ analyzed the data. YL, MS, PL, and XZ interpreted the data. MS and YL wrote the article. YL and MS obtained research funding. All authors read and approved the final manuscript.
The authors declare that they have no competing interests.
Consent for publication
Patients provided written consent for the use of their images and video.
Ethics approval and consent to participate
This article does not contain any studies with human participants performed by any of the authors. All procedures performed in studies involving patients were in accordance with the ethical standards of the Lanzhou General Hospital of PLA Animal Care Committee with the number 2013-GKJ-048 or practice at which the studies were conducted.
Open AccessThis article is distributed under the terms of the Creative Commons Attribution 4.0 International License (http://creativecommons.org/licenses/by/4.0/), which permits unrestricted use, distribution, and reproduction in any medium, provided you give appropriate credit to the original author(s) and the source, provide a link to the Creative Commons license, and indicate if changes were made. The Creative Commons Public Domain Dedication waiver (http://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated.
- Kaufman MR, Miller TA, Huang C, Roostaeian J, Wasson KL, Ashley RK, et al. Autologous fat transfer for facial recontouring: is there science behind the art? Plast Reconstr Surg. 2007;119(7):2287–96.Google Scholar
- Kaufman MR, Bradley JP, Dickinson B, Heller JB, Wasson K, O’Hara C, et al. Autologous fat transfer national consensus survey: trends in techniques for harvest, preparation, and application, and perception of short- and long-term results. Plast Reconstr Surg. 2007;119(1):323–31.View ArticlePubMedGoogle Scholar
- Carpaneda CA, Ribeiro MT. Study of the histologic alterations and viability of the adipose graft in humans. Aesthetic Plast Surg. 1993;17(1):43–7.View ArticlePubMedGoogle Scholar
- Carpaneda CA, Ribeiro MT. Percentage of graft viability versus injected volume in adipose autotransplants. Aesthetic Plast Surg. 1994;18(1):17–9.View ArticlePubMedGoogle Scholar
- Coleman SR. Structural fat grafting: more than a permanent filler. Plast Reconstr Surg. 2006;118(3 Suppl):108s–20s.Google Scholar
- Meier JD, Glasgold RA, Glasgold MJ. Autologous fat grafting: long-term evidence of its efficacy in midfacial rejuvenation. Arch Facial Plast Surg. 2009;11(1):24–8.View ArticlePubMedGoogle Scholar
- Mei S, Yi L, Zheng W, Ping L, Bin X, Xianying Z. The development and clinical application of facial minimally invasive fat granules injection device with controlled pressure and quantity. Zhonghua Zheng Xing Wai Ke Za Zhi. 2015;31(6):432–6.PubMedGoogle Scholar
- Mojallal A, Shipkov C, Braye F, Breton P, Foyatier JL. Influence of the recipient site on the outcomes of fat grafting in facial reconstructive surgery. Plast Reconstr Surg. 2009;124(2):471–83.View ArticlePubMedGoogle Scholar
- Viard R, Bouguila J, Voulliaume D, Comparin JP, Dionyssopoulos A, Foyatier JL. Fat grafting in facial burns sequelae. Ann Chir Plast Esthet. 2012;57(3): 217–29.View ArticlePubMedGoogle Scholar
- Gamboa, GM and Ross WA. Autologous fat transfer in aesthetic facial recontouring. Ann Plast Surg. 2013;70(5):513–6.View ArticlePubMedGoogle Scholar
- Calabria R, Hills B. Fat grafting: fact or fiction? Aesthet Surg J. 2005;25(1):55.View ArticlePubMedGoogle Scholar
- Gonzalez AM, Lobocki C, Kelly CP, Jackson IT. An alternative method for harvest and processing fat grafts: an in vitro study of cell viability and survival. Plast Reconstr Surg. 2007;120(1):285–94.View ArticlePubMedGoogle Scholar
- Barret JP, Sarobe N, Grande N, Vila D, Palacin JM. Maximizing results for lipofilling in facial reconstruction. Clin Plast Surg. 2009;36(3):487–92.View ArticlePubMedGoogle Scholar
- Xie Y, Zheng DN, Li QF, Gu B, Liu K, Shen GX, et al. An integrated fat grafting technique for cosmetic facial contouring. J Plast Reconstr Aesthet Surg. 2010;63(2):270–6.Google Scholar
- Asken S. Facial liposuction and microlipoinjection. J Dermatol Surg Oncol. 1988;14(3):297–305.View ArticlePubMedGoogle Scholar
- Nguyen A, Pasyk KA, Bouvier TN, Hassett CA, Argenta LC. Comparative study of survival of autologous adipose tissue taken and transplanted by different techniques. Plast Reconstr Surg. 1990;85(3):378-86. discussion 387–389.View ArticlePubMedGoogle Scholar
- Rusciani Scorza A, Rusciani Scorza L, Troccola A, Micci DM, Rauso R, Curinga G. Autologous fat transfer for face rejuvenation with tumescent technique fat harvesting and saline washing: a report of 215 cases. Dermatology. 2012;224(3):244–50.View ArticlePubMedGoogle Scholar
- Pu LL. Towards more rationalized approach to autologous fat grafting. J Plast Reconstr Aesthet Surg. 2012;65(4):413–9.View ArticlePubMedGoogle Scholar
- Nguyen PS, Desouches C, Gay AM, Hautier A, Magalon G. Development of micro-injection as an innovative autologous fat graft technique: The use of adipose tissue as dermal filler. J Plast Reconstr Aesthet Surg. 2012;65(12):1692–9.View ArticlePubMedGoogle Scholar
- Morán MJ, Navarro I, Sánchez R, Burgueño M. Use of a three-way stopcock in grafting of autologous fat: technical note. Br J Oral Maxillofac Surg. 2012;50(5):474–5.View ArticlePubMedGoogle Scholar
- Fulton JE, Parastouk N. Fat grafting. Dermatol Clin. 2001;19(3):523–30.View ArticlePubMedGoogle Scholar
- Yoshimura K, Sato K, Aoi N, Kurita M, Hirohi T, Harii K. Cell-assisted lipotransfer for cosmetic breast augmentation: supportive use of adipose-derived stem/stromal cells. Aesthetic Plast Surg. 2008;32(1):48-55. discussion 56–57.View ArticlePubMedGoogle Scholar
- Hetherington HE, Block JE. Facial fat grafting with a prototype injection control device. Clin Cosmet Investig Dermatol. 2013;6:201–9.PubMedPubMed CentralGoogle Scholar
- Lin TM, Lin TY, Chou CK, Lai CS, Lin SD. Application of microautologous fat transplantation in the correction of sunken upper eyelid. Plast Reconstr Surg Glob Open. 2014;2(11):e259.View ArticlePubMedPubMed CentralGoogle Scholar